In all other areas, products must be returned to a service facility designated by Agilent Technologies. Use this software only with the A. Power Supply recovery time is adjustable from 0 to msec. Agilent Technologies makes no warranty of any kind with regard to this material, including, but not limited to, the implied warranties of merchantability and fitness for a particular purpose. Refer to Figure on page for a detailed DC Schematic. Fill out a service tag and attach it to the instrument. Never use styrene pellets in any shape as packaging materials.
|Date Added:||12 December 2009|
|File Size:||35.23 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Accessories Firmware Revisions Table atyenuator If the shipping container is damaged, or the cushioning material shows signs of stress, notify the carrier as well as Agilent Technologies. Outside Agilent Technologies service travel areas, warranty service will be performed at Buyer s facility only upon Agilent Technologies prior agreement and Buyer shall pay Agilent Technologies round trip travel expenses.
The power supply allows the switch driver four switching operations at the same time.
Seal the shipping container securely with strong nylon adhesive tape. Switch one, wired to J1 on driver card 1, would have a channel address of The default setting is 30 ms.
HP – Agilent – Keysight A Attenuator, Switch Driver
These low impedance pulses can be adjusted for pulse width in order to optimize switching speed. Event Status Register Query Table attenkator It calls attention to an operating procedure, practice, or the like that, if not correctly performed or adhered to, could result in personal injury or death.
Mark the shipping container Fragile, Handle with Care to help ensure careful handling.
Swjtch Agilent Technologies warrants that each Agilent Technologies hardware, software, and firmware product on Agilent Technologies Corporate Price List dated July 1, or later delivered under the product s contract of sale will be able to accurately process date data including, but not limited to, calculating, comparing, and sequencing from, into, and between the twentieth and twenty-first centuries, and the years andincluding leap year calculations, when used in accordance with the product documentation provided that all other products that is, hardware, software, firmware used in combination with such Agilent Technologies product s properly exchange date data with it.
Warranty service for products installed by Agilent Technologies and certain other products designated by Agilent Technologies will be performed at Buyer s facility at no charge within Agilent Technologies service travel areas.
HP/AGILENT A ATTENUATOR/SWITCH DRIVER
This results in a default switching speed attenuaor 0. WIDTh on page Determining Your Module Serial Number Environmental Limitations A serial number is attached to a label on the rear switxh of the module. NOTE The channel number must be preceded by the driver card number. Used to indicate a variable: If you do not have access to the Internet, one of these centers can direct you to your nearest representative: Power Supply recovery time is adjustable from 0 to msec.
Showing Results for Keysight Technologies (Agilent HP) 87130A
Each channel has its own unique address. The controller assembly actuates the transistor for a predetermined time that is set by the Width command. Thus, if each switch has ma actuating plus 50 ma quiescent current, a combined length of The remaining digits are assigned sequentially and are different for each instrument.
Power cord In accordance with international safety standards, a three-wire power cable is provided with this instrument.
Keysight Technologies 87130A FOR RENT
Refer to Service and Support on page v for ordering information. Refer to Figure This document contains proprietary information that is protected by copyright.
You can find a list of local service representatives on the Web at: Nothing in this warranty will be construed to limit any rights or remedies provided elsewhere in the contract of sale with respect to matters other than year compliance.
Personnel should be grounded with a switvh wrist strap before touching the center pin of any connector and before removing any assembly from the instrument. Using the Internal Driver 1.
S1 up is open or away from the PC board. Alternately, an externally installed switch or circuit breaker, which is readily identifiable and is easily reached by the operator, may be used as a disconnecting device. The internal driver card is set to card atrenuator. Agilent Technologies makes no warranty of any kind with regard to this material, including, but not limited to, the implied warranties of merchantability and fitness for a particular purpose.